Individual standard medical packing,
EO sterilized with net weight 10 grams.
6 pieces per box of L25 x W16 x H6.50 cm
Net weight 250 gram gross weight 260 grams
20 boxes per carton of L52 x W32 x H34 cm (0.0566 CBM)
20 days up
Air cushion compression artery tourniquet
CE approved tourniquet
air cushion type
Medical Supply Tourniquet
Disposable Tourniquet- Manual
Product name: Disposable Tourniquet
Model types:Tourniquet for radial artery, simple tourniquet, tourniquet
For femoral artery
1. Transparent design enhances the precise observationof blood compression in the artery puncturing spot;
2. It can adjust pressure by tighten or loosen the screw nut to enhances and make postoperative blood compression much easier to operate;
3. It will ensure the accuracy of pressing at the puncture spot, while not pressing the ulnar artery in the meantime;
4. Since the pressing rubber pad is made of special TPE material in which fixing patient’s body is not necessary and it will alleviate the pain of the patient to a great extent;
5. Toavoid the stanching part from bleeding again.
Main components: consisted of elastic band for radial artery tourniquet, self-adhesive tape, bracket, compressing mat and rubber mat.
Femoral artery tourniquet made of 1.8m long strap, self-adhesive tape, silicone mat, supporter, striped compressing pad and bolt.
Scope of application: for bleeding stanching after radial artery or femoral artery punctures
Tourniquet for radial artery:
1.After the puncturing of the radial artery, use a sterilized gauze to compress the puncture site;
2.Let the bolt of the tourniquet pass through the bolt hole of the bracket, wrap around the patient wrist;
3.Compress the mat on the artery, place the soft side of the compressing mat on the gauze;
4. Adjust the elastic band to the medium pressure.
Tourniquet for femoral artery:
1.After the puncturing of the femoral artery, use a sterilized gauze to compress the puncture site;
2.Hold the tourniquet; apply the central point of the compressing mat on the area 1-3 cm above the puncture site. Extend the elastic band along the groin towards the inner side of the thigh and wind to the rear buttock, through the area beneath the anterior superior iliac spine. At last, pass the movable end of the elastic band through the side hole on the fixing end of the compressing mat and fasten it;
3.Swirl the handle of the bolt on the compressing mat; let the soft side of the mat compress tightly on the puncture site. Adjust the depth of swirling, normally touch the pulse of dorsal artery of foot as a standard (just for reference)
4.Use one hand to compress the bolt on the artery, while using another hand to swirl the top of bolt nut on the bracket to fix the location;
5.Observe whether the puncture site is bleeding, which helps to adjust the compression force at once;
6.Use the fixable assist strap: stick the tip of the adhesive tape (double-sided adhesive) on the compressing band of the femoral artery (the outer side of the thigh), wind back to the contralateral anterior superior iliac spine and tighten. As to the other tip, repeatedly fix on the tip of the adhesive tape.
1.This product is for single use only, destroy it after use;
2.Observe whether the puncture site is bleeding while applying the product, inspect the compression force, avoid bleeding and tissue necrosis;
3.This product has been sterilized by Ethylene Oxide, if the package is broken, must take back to the manufacturer to sterilize again before use;
4.The sterilization period lasts 3 years, don’t use after expired.
Storage:Avoid sunlight, store in dry, in room temperature.